YY/T 1288-2015 Translated English of Chinese Standard. (YYT 1288-2015, YY/T1288-2015, YYT1288-2015): Nylon blood filter nets for transfusion equipment for single use
https://www.chinesestandard.net, 2018 M04 23 - 14 páginas
This Standard specifies the requirements for nylon blood filter nets for transfusion equipment for single use (hereinafter referred to as the filter net). The filter net can be installed in the dropping funnel or accessories of transfusion (blood collection) apparatus, used to filter blood clots, impurities and foreign bodies in blood or blood products.
Annex A Normative Test method of filter net particle pollution
Annex B Normative Test method of filtration performance of filter net
45mm is marked aperture of 0.8 bag filter net batch number beaker blank control solution blood filter nets bracket length China 3 Volumes corrected vision eluent Example of typical exceed 10 particles/mL filter is organized filter net length filter net structure filternet flow the test Flush the fluid GB/T grid membrane grille membrane length is 45mm manufacturer's name and/or mass concentration measuring cylinder melting peak temperature membrane filtration method to flow monofilament melting peak name and/or trademark nets for transfusion non-contaminating method number of particles nylon 1010 monofilament nylon 66 Nylon blood filter organization of filter Particle counter particle size particle test solution peak temperature melting Pharmacopoeia port width product effective date retention rate rinse solution B.3 sample single warp single Standard test liquid Test method test solution B.2 tested according transfusion direction type inspection typical bag filter typical bracket filter visible foreign matter warp single weft Welding firmness Zhang